■ Installation Conditions (main items)
Single phase AC200V 25A or more
Water supply facility
Connect the equipment to a water supply facility that has a water pressure of 0.23MPa to 0.40MPa, flow rate of 6 L / min
※Check that the water pressure does not fall below the specified value when using another water supply system.
If this condition is not met, a pressure pump may be required.
A drain port is 3 / 8 inch female thread specification.
Connect the equipment to a heat-resistant drainage facility as hot water may be discharged. (Steel pipes recommended)
Install the equipment in a horizontal place where water does not splash.
When installing the equipment, make a space of 20 cm or more on the left and right sides and the rear side.
Avoid installing the equipment in a place where the room temperature becomes 5℃ or less, or 35℃ or more.
A boiler is not used.